Home > News
Knowledge of Jingkangyu Medical Nebulizer
Please keep the best condition of the product and follow the following seven precautions to avoid damage to the product:
1. Do not place the product in the place where infants or unconscious patients touch.
2. Do not subject the product to any strong impact, such as dropping the product from a height.
3. When there is liquid in the liquid bottle, please do not carry or store the product.
4. Do not place the product in places with high temperature, low temperature, high pressure or direct sunlight.
5. Don't fold the trachea.
6. For the disposal methods of the discarded body, accessories and optional parts, please follow the regulations of the local government.
7. Do not use benzene, thinner and flammable chemicals to clean the product.
② Second, replace the filter
When the filter changes color, please replace it with a new one. The operation steps are as follows.
1. Pull out the filter cover and remove it, then you can replace the filter.
2. Use sharp tools such as toothpicks to remove the filter. Please discard the removed filter and replace it with a new one.
3. Install the filter cover.
Only use filters specially designed for this product. Do not use this product without a filter.
The filter has no distinction between front and back.
Before replacement, make sure the filter is clean and free of dust.
Do not clean the filter. If the filter is wet, replace the filter. The filter will become clogged when wet.
To prevent clogging of the filter cover, clean the filter cover regularly. Do not boil the filter cover. When cleaning, be sure to remove the filter cover first. After washing, fully dry, then install the filter cover according to the correct method.
Related searches: medical sprayers, medical sprayers, medical sprayer sales, wholesale masks, medical equipment sprayer companies
Address：Jiangxi Ganzhou Dingnan Taihu Industrial area